|Year : 2019 | Volume
| Issue : 2 | Page : 114-122
|Effectiveness of rotary and reciprocating systems on microbial reduction: A systematic review
Riluwan Siddique, Malli Sureshbabu Nivedhitha
Department of Conservative Dentistry and Endodontics, Saveetha Dental College, Saveetha Institute of Medical and Technical Sciences, Saveetha University, Chennai, Tamil Nadu, India
Click here for correspondence address and email
|Date of Submission||29-Nov-2018|
|Date of Decision||16-Jan-2019|
|Date of Acceptance||29-Jan-2019|
|Date of Web Publication||02-May-2019|
| Abstract|| |
Introduction: The role of microorganism and their products in the initiation, propagation, and persistence of periradicular periodontitis has been established. One of the major goals of the treatment of infected root canals of teeth with apical periodontitis is to promote maximal reduction in the intracanal bacterial populations. Engine-driven nickel-titanium instruments possess the latest generation of root canal instruments. The possible benefit of rotary instrumentation over other instrumentation techniques regarding cleaning and disinfecting effects would be irrigant warming and/or turbulence caused by the mechanical rotation of instruments. Furthermore, reciprocating instrument has been introduced for root canal preparation. It has been shown that instruments subjected to reciprocation have increased resistance to fatigue and longer usual life when combined with instruments used in continuous rotation motion. The reciprocating system uses single-file instrumentation technique which can shape and clean the canal in a shorter period and together with the lesser amount of antimicrobial agent.
Objective: The objective of this study is to compare and evaluate the microbial reduction of rotary and reciprocating systems on microbial reduction.
Search Strategy: A search was performed in Electronic databases (i.e., PubMed, Cochrane library, Science direct, Lilac, Sigle) using following search terms alone and in combination by means of PubMed search builder from January 1985 to December 2017.
Selection Criteria: Studies were selected if they met the following criteria: In vivo studies comparing rotary and reciprocating system in asymptomatic apical periodontitis patients.
Main Results: The results showed that the reciprocating system exerted an almost similar antibacterial effect when compared with the rotary system.
Conclusion: The present systematic review does not provide concrete evidence to show increased antibacterial efficacy of reciprocating system as compared to the rotary system. Furthermore, clinical trials are required to evaluate the efficacy of various instrumentation systems in reducing bacteria from the root canal system.
Keywords: Apical periodontitis; microbial reduction; reciprocating files; rotary files
|How to cite this article:|
Siddique R, Nivedhitha MS. Effectiveness of rotary and reciprocating systems on microbial reduction: A systematic review. J Conserv Dent 2019;22:114-22
|How to cite this URL:|
Siddique R, Nivedhitha MS. Effectiveness of rotary and reciprocating systems on microbial reduction: A systematic review. J Conserv Dent [serial online] 2019 [cited 2023 Oct 4];22:114-22. Available from: https://www.jcd.org.in/text.asp?2019/22/2/114/257587
| Introduction|| |
Apical periodontitis is a disease caused by bacterial infection of the root canal system which can be controlled by proper endodontic treatment., Removing pulp tissue remnants, microorganism, and microbial toxin from root canal system is necessary for the success of root canal treatment. The main goal of root canal treatment is to reduce the amount of bacteria and their by-products which contributes to the perpetuation of apical periodontitis.,, Lipopolysaccharides are most important by-products present on the outer layer of Gram-negative bacterial species.,, Root canals can be cleaned by instrumentation along with irrigants and intracanal medicaments. To make canal preparation easier, several mechanical devices and techniques were introduced to further improve the effectiveness of instrumentation. The devices may be classified as either manual or machine-assisted for root canal instrumentation. Machine-assisted endodontic instruments help in easier and faster instrumentation. A machine-assisted technique includes automated root canal preparation, sonic and ultrasonic preparation, laser systems, and non-instrumental techniques.
Stainless steel instruments been used in previous years have showed more chances of canal transportation. They have become dispensable with the introduction of the more flexible nickel-titanium (NiTi) instruments. With the introduction of M-wire or control memory wire improved the instruments (easier, faster and better root canal shaping, greater resistance to fracture).
According to Hulsmann et al. (2005), Oltramare (1892) was first to give reference to rotary instrumentation, who used fine needles with rectangular cross sections that could be attached to a dental handpiece. The endodontic handpiece was first introduced by Rollins (1899) for root canal instrumentation that was used with specially designed needles at 100 rpm (Milas 1987). The use of stainless-steel instruments was limited after the introduction of NiTi endodontic hand instruments by Walia et al. (1988) thereby avoided canal transportation. Recently, Diemer et al. 2013 have introduced a new type of rotary motion called asymmetrical rotary motion. Asymmetrical rotary motion (waves of motion traveling along the active part of the file) includes off-centered cross-section which is produced by design features of the file, that is not related to endodontic motors. The new systems come into practice as asymmetrical motion were Revo-S (Micro-Mega, Besancon, France), followed by ProTaper Next (DENTSPLY Tulsa Dental Specialties), new generation OneShape (Micro-Mega) and TRUShape 3D Conforming Files (DENTSPLY Tulsa Dental Specialties).
Endodontic instruments were initially manufactured using carbon steel. Their strength and quality vastly increased with the introduction of stainless steel. In the late 1980s, Ni–Ti alloy took a major change in endodontic Instruments. Ni–Ti alloy confers two distinct advantages: superelasticity and high resistance to cyclic fatigue. The crown-down technique is the key factor contributing to the backbone of Ni–Ti rotary instrumentation, which involves three phases as follows: coronal access, coronal-middle preparation, and apical preparation. Before commencement of large rotary files used to enlarge the canal short of the apex before smaller and smaller files are introduced until full working length can be achieved. The apical third has been enlarged to the desired size using larger files. Most other techniques that are used with continuous rotation system are simply variations of the crown-down.
Variety of different endodontic filing systems is available on the dental materials market today. Most of these tapered files which follow ISO standards are compatible with the crown-down instrumentation technique. Some of them include Profile, Vortex, GT, EndoSequence, K3, HERO, Twisted File, and Hy-Flex. The ProTaper Universal system has unique where its taper varies over the course of its long axis. Dr. Cliff Ruddle, Dr. John West, and Dr. Pierre Machtou developed the system in the mid-1990s. It comprises 7 files: 2 shaping files and 5 finishing files. Each Protaper Universal file is capable of working to full length. A wider diameter of the shaping files takes care of coronal-middle preparation phase and the desired apical size determines the number of finishing files in the sequence that are used.
The ProTaper Next has introduced into the market with expectations of better canal shaping and removal of bacteria from root canal walls. It has rectangular cross-section unlike ProTaper universal (convex triangular) cross-section which is off-centered. The axis of rotation of ProTaper universal is different than its center of mass and ProTaper universal limits the amount of surface area in contact between the file and canal wall. Protaper Next includes five files and is designed to attain full apical preparation with fewer Instruments while producing slightly less tapered sizes. The files, X1–X5, have apical Sizes of 0.17 mm, 0.25 mm, 0.30 mm, 0.40 mm, and 0.50 mm and tapers of 0.04 mm, 0.06 mm, 0.07 mm, 0.06 mm, and 0.06 mm, respectively. In comparison to ProTaper Universal, ProTaper Next files have been found to possess significantly enhanced performance with respect to cyclic fatigue resistance and in minimizing extruded debris, microcrack induction, and canal transportation.
Various endodontic reciprocating handpieces have been developed (Prichard 2012) after reciprocation was first introduced in 1964 with the Giromatic system (Micro-Mega). The Giromatic system, Endo-Gripper (Moyco Union Broach, Montgomeryville, PA, USA), Intra-Endo 3 LD (KaVo, Biberach, Germany) and Dynatrak (DENTSPLY DeTrey, Konstanz, Germany) which seems to operate with equal angles of 90° clockwise (CW) and counter CW (CCW) motion. Since Giromatic system produced greater procedural errors than hand filing, it lost popularity gradually.
The M4 (SybronEndo, Orange, CA, USA), Endo-Eze (Ultradent Products Inc., South Jordan, UT, USA) and Endo-Express SafeSider (Essential Dental Systems, South Hackensack, NJ, USA) systems are recent files which use reciprocating handpieces that utilize 30° angles of CW and CCW rotation. These handpieces enable the formation of an endodontic glide path using small stainless steel hand files.
The new era of NiTi alloys regained popularity of reciprocating systems and endodontic torque control motors. Roane et al. described a balanced force technique for curved canals and included unequal CW and CCW motions with hand file. The concept of single-file reciprocation was introduced by Yared which was based on a balanced-force technique and used the ProTaper F2 instrument (DENTSPLY Tulsa Dental Specialties) (a flute-designed instrument with cross-sectional geometry that aids in the cutting of the dentine in the CW direction) with unequal CW and CCW rotational motion (144° CW and 72° CCW). This development points to the fact that the instruments required five rotations to complete a full 360° rotation. At the same time, due to this motion, the elastic limit of the instrument was not exceeded.
The manufacturers introduced single-file reciprocating systems including WaveOne (DENTSPLY Tulsa Dental Specialties) and Reciproc (VDW, Munich, Germany) based on developments in motions. In case if instruments have CCW cutting direction, it would cut when the CCW movement was greater than the CW movement. However, all of the instruments are designed for cutting in the CW direction except for reciprocating instruments. There are more chances of instrument will neither cut nor penetrate into the canal if one tries to use CW cutting instruments with the reciprocating motors where CCW motion is greater than CW motion.
In the majority of cases, a single file is sufficient to complete root canal preparation in single-or multiple root canal systems. The files work in a reverse “balanced force” cutting motion which is driven by a preprogrammed motor (X-Smart Plus motor fitted with 6:1 reducing hand piece) (DENTSPLY/Maillefer) that is capable of turning the files in a back and forth “reciprocating” motion instead of a rotary motion, The CCW movement of 150 degrees is capable of advancing the instrument apically as the dentine on the root canal wall is engaged and cut. This is followed by 30° CW movement thereby ensures that the instrument disengages before excessive torsional stress is transferred onto the metal alloy and before the instrument can bind (taper lock) into the root canal. CCW rotation allows the instrument to advance apically into the root canal.
Reciprocating endodontic motors can be open or closed motors, where open motors ([ATR Vision; ATR, Pistoia, Italy], [iEndo Dual; Acteon, Merignac, France], [SAF pro System; ReDent-Nova, Ra'nana, Israel]) allows modification of angles and speed, whereas closed motors ([WaveOne], [Reciproc], [Elements motor; SybronEndo], [ATR Technika; ATR]) were not allowed such modifications.
Reciprocating systems provides greater comfort for both patient and professional as it reduces working time, making it four times faster than the traditional rotary file system. Its a matter of debate regarding the ability of single-file system to shape and clean the canal in shorter period and together with the lesser amount of antimicrobial agent when compared with multiple file rotary system. According to Alves FR et al. 2012, single-file reciprocating instrumentation technique showed similar intracanal bacterial reduction when compared with rotary instrumentation if the irrigant volume and the time of irrigation are similar.
The aim of this systematic review was to assess the microbial reduction of rotary and reciprocating systems in asymptomatic apical periodontitis patients.
Does the reciprocating system bring about significant removal of microbes from root canal space compared to rotary systems?
There is no significant difference between the anti-bacterial effectiveness of rotary and reciprocating system in microbial reduction.
There is a significant difference between the anti-bacterial effectiveness of the rotary and reciprocating system in microbial reduction.
The PICO question can be framed as follows:
- Population - patients with asymptomatic apical periodontitis
- Intervention – reciprocating system
- Comparison – rotary system
- Outcome – microbial reduction.
| Materials and Methods|| |
The present systematic review was carried out following approval of Institutional Review Board (SRB/MDS/ENDO/17-18/0025).
Detailed search strategies were developed for the database searched for identification of studies included or considered for this review.
- Cochrane Library (CENTRAL)
- Science Direct
There were no language restrictions.
The following journals were hand searched:
- International Endodontic Journal
- Journal of Endodontics.
Criteria for considering studies for this review:
Types of participants:
Teeth with asymptomatic apical periodontitis
Types of interventions:
Comparison– Rotary systems.
Types of outcome measures:
Types of studies:
In vivo studies.
The following studies were excluded,
- In vitro studies
- Animal studies
- Ex vivo studies
| Results|| |
The search identified 520 publications, out of which 17 duplicate articles and 481 articles were excluded after the title search. Full articles were obtained for 8 studies and evaluated. After reading the abstract and full-text article, 5 of these were excluded. Therefore, a total of three publications fulfilled all criteria for inclusion. [Table 1] shows the search methodology. Some of the characteristics of excluded articles are given in [Table 2]. [Figure 1] shows the PRISMA flow diagram and general information of the selected articles are given in [Table 3].
|Table 3: General information of variables of interest and interpretation|
Click here to view
| Discussion|| |
The results of this systematic review confirm that more evidence-based research in this area is needed and that there have been limited randomized controlled clinical trials in this area. With the limited information currently available, the best current available clinical evidence suggests that both rotary and reciprocating files exerted almost similar antibacterial effectiveness.
It has been proved that bacteriological conditions of root canal before obturation may influence the long-term prognosis of teeth, with significantly better results with negative cultures. Clinical studies should be carried out to correlate culture-independent molecular data with treatment outcome. It is clearly well-known fact that maximum bacterial elimination is required for long-term prognosis. Studies comparing the long-term outcome after treatment with these instruments should be prompted as they have a long life in the market.
Bacterial reduction in root canals was found significant when compared Reciproc R40 and BioRaCe under the same irrigation protocol. In another study, significant count reductions of Enterococcus faecalis WaveOne and Reciproc were used. Investigators compared three different systems namely Reciproc R25, SAF, and TF systems to check for bacterial reduction in mesial roots of mandibular molars. Results showed that the ability of all systems to promote significant bacterial reduction despite the detection of positive cultures which may be due to the complexity of mesial root canal systems. A clinical study showed that both reciprocating files and rotary full sequence system showed similar effectiveness in reducing endotoxins and cultivable bacteria, but they were not able to completely disinfect all root canal systems analyzed. These results show that the mechanical action of endodontic files on dentin together with adequate exposure to NaOCl irrigation is more effective for disinfection than the number of files used.
Many studies have proved that single-file reciprocating systems and full-sequence rotary systems have similar effect on microbial reduction (Siqueira et al. 2013, Marinho et al. 2015, Martinho et al. 2014, Neves et al.2016).,,,, Marinho et al. (2015) showed that Reciproc, Mtwo, ProTaper, and Race (FKG Dentaire, La Chaux-de-Fonds, Switzerland) instruments produced highly significant reductions of the bacterial load, but no significant differences in endotoxin levels were found. However, there are limited data concerning the effect of pure reciprocating motion on bacterial reduction.
Reciproc and BioRaCe have showed significant bacterial reductions despite some number of bacteria were still detected by qPCR in 55% and 50% of the canals. This is in agreement with few studies that used different instrumentation techniques and NaOCl irrigation.,, This bacterial count may still be sufficient to maintain the disease. Bacteria may remain in root canal system due to canal morphology such as isthmi, lateral canals, and apical ramifications which may not be reached by instruments and irrigants., This can be avoided by enlarging canal enough to incorporate bacterial walls in the final preparation. Most of the instrument systems are available in predetermined size series which creates a problem in removing canal bacteria, whereas self-adjusting file system follow root canal morphology, thereby helps in bacterial elimination.
Single-file techniques have been used from start to finish because of its convenience rather than clinically proven effectiveness.,,, Systems with greater number of instruments along with the higher frequency of irrigation during biomechanical preparation reduce microorganisms and endotoxins more likely than single-file system. According to Kakehashi et al., the presence of bacteria is the main cause of the development of periodontal infection and apical periodontitis. The aim of chemo-mechanical root canal preparation is to eradicate residual vital and non-vital pulp tissue and reduce the number of pathogenic organisms. Mechanical instrumentation can disrupt the bacterial biofilm and has the potential to reduce the presence of bacteria in the main root canal. The rotary systems, ProTaper Universal and Mtwo have been shown to provide adequate geometry and substantial bacterial reduction in the root canal.
A recent invitro study compared the influence of the reciprocating single-file technique with conventional rotary instrumentation on the microbial reduction in infected root canals. In their study, they have used WaveOne, Reciproc, ProTaper Universal, Mtwo, and manual instrumentation. The results demonstrated that the bacterial count was significantly reduced after instrumentation in all groups. However, there were no significant differences in the bacterial count reductions effected by the reciprocating, rotary, and manual techniques. The study concluded that reciprocating systems resulted in similar bacterial reductions to those obtained with rotary systems or with the manual instrumentation technique. Similar observations were made by Nabeshima et al. in 2014.
The purpose of this review is to evaluate the antibacterial efficacy of the rotary and reciprocating system in microbial reduction. Three clinical studies fulfilled the criteria for being included in this review (Monica A. S. Neves et al. 2016, DaianaCavalli et al. 2017, Frederico C. Martinho et al., 2014).
A total number of 137 participants were included in this review. Out of 137 participants, 73 subjects received reciprocating instrumentation and 64 participants received rotary instrumentation. Age group of the participants was 16–85 years.
These three randomized controlled clinical trials, all canals were completely instrumented in a single visit. Bacterial load in root canal samples of teeth were checked before (S1) and after (S2) chemo mechanical instrumentation using reciprocating single instrument or a continuously rotating multi-instrument series.
Mostly systematic reviews will perform meta-analysis, which involves the statistical pooling of data from individual studies when the studies are similar. A meta-analysis can yield a more precise overall estimate of the treatment effect. However, meta-analysis may not be appropriate in many situations. Due to the heterogeneity among the studies such as difference in the composite types, sample sizes and follow-up periods, we could not perform a meta-analysis to summarize the data of included studies. Hence, only descriptive evaluation of data has been provided.
Report on quality of evidence looked upon
All 3 studies included in this review have a level of evidence 3(OCEBM, 2011). Thus, the level of evidence is moderate. Risk of bias of 3 studies (Monica A. S. Neves et al. 2016, DaianaCavalli et al. 2017, Frederico C. Martinho et al. 2014) had a high risk of bias. The risk of bias summary and graph [Figure 2] and [Figure 3] were generated using Review Manager 5 software (RevMan 5).
From this systematic review, it can be concluded that microorganism is more likely to be reduced when systems with a greater number of instruments are used.
Implications for practice
Single-file reciprocating systems was found almost similar microbial reduction when compared with rotary systems.
Implications for research
Since the number of randomized controlled clinical trials included in this review is limited, more clinical trials are required to prove the efficacy of various instrumentation systems.
Report of outlier data
No outlier data obtained.
| Summary|| |
The aim of this systematic review was to evaluate the effectiveness of rotary and reciprocating systems in removing bacteria from root canal space. An electronic search was carried out on PUBMED database for the articles which could be used for evaluating the effectiveness of reciprocating system in removing bacteria from root canal when compared with rotary systems.
Article search were selected based up on the pre-stated inclusion and exclusion criteria. A total of three articles were included in this systematic review for detailed evaluation.
The microbial reduction of rotary and reciprocating systems was the primary variable of interest.
Based on the result of this systematic review, we can conclude that single-file reciprocating system exerted almost similar microbial reduction when compared with rotary systems.
| Conclusion|| |
The present systematic review does not provide concrete evidence to show increased antibacterial efficacy of reciprocating system as compared to rotary system. Three articles included in this review have shown a high risk of bias. Further, it also recommends more clinical studies to be done comparing different rotary and reciprocating systems on microbial reduction.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Sjögren U, Figdor D, Persson S, Sundqvist G. Influence of infection at the time of root filling on the outcome of endodontic treatment of teeth with apical periodontitis. Int Endod J 1997;30:297-306.
Fabricius L, Dahlén G, Sundqvist G, Happonen RP, Möller AJ. Influence of residual bacteria on periapical tissue healing after chemomechanical treatment and root filling of experimentally infected monkey teeth. Eur J Oral Sci 2006;114:278-85.
Basmadjian-Charles CL, Farge P, Bourgeois DM, Lebrun T. Factors influencing the long-term results of endodontic treatment: A review of the literature. Int Dent J 2002;52:81-6.
Martinho FC, Chiesa WM, Marinho AC, Zaia AA, Ferraz CC, Almeida JF, et al.
Clinical investigation of the efficacy of chemomechanical preparation with rotary nickel-titanium files for removal of endotoxin from primarily infected root canals. J Endod 2010;36:1766-9.
Alves FR, Rôças IN, Almeida BM, Neves MA, Zoffoli J, Siqueira JF Jr., et al.
Quantitative molecular and culture analyses of bacterial elimination in oval-shaped root canals by a single-file instrumentation technique. Int Endod J 2012;45:871-7.
Machado ME, Nabeshima CK, Leonardo MF, Reis FA, Britto ML, Cai S, et al.
Influence of reciprocating single-file and rotary instrumentation on bacterial reduction on infected root canals. Int Endod J 2013;46:1083-7.
Schein B, Schilder H. Endotoxin content in endodontically involved teeth 1975. J Endod 2006;32:293-5.
Pitts DL, Williams BL, Morton TH Jr. Investigation of the role of endotoxin in periapical inflammation. J Endod 1982;8:10-8.
Nair PN. Pathogenesis of apical periodontitis and the causes of endodontic failures. Crit Rev Oral Biol Med 2004;15:348-81.
Bystrom A, Sundqvist G. The antibacterial action of sodium hypochlorite and EDTA in 60 cases of endodontic therapy. Int Endod J 1985;18:35-40.
Schäfer E, Schulz-Bongert U, Tulus G. Comparison of hand stainless steel and nickel titanium rotary instrumentation: A clinical study. J Endod 2004;30:432-5.
Hülsmann M, Peters OA, Dummer PM. Mechanical preparation of root canals: Shaping goals, techniques and means. Endod Topics 2005;10:30-76.
Cheung GS, Liu CS. A retrospective study of endodontic treatment outcome between nickel-titanium rotary and stainless steel hand filing techniques. J Endod 2009;35:938-43.
Walia HM, Brantley WA, Gerstein H. An initial investigation of the bending and torsional properties of nitinol root canal files. J Endod 1988;14:346-51.
Diemer F, Michetti J, Mallet JP, Piquet R. Effect of asymmetry on the behavior of prototype rotary triple helix root canal instruments. J Endod 2013;39:829-32.
Prichard J. Rotation or reciprocation: A contemporary look at NiTi instruments? Br Dent J 2012;212:345-6.
Weine FS, Kelly RF, Bray KE. Effect of preparation with endodontic handpieces on original canal shape. J Endod 1976;2:298-303.
Gambarini G, Plotino G, Sannino G, Grande NM, Giansiracusa A, Piasecki L, et al.
Cyclic fatigue of instruments for endodontic glide path. Odontology 2015;103:56-60.
Roane JB, Sabala CL, Duncanson MG Jr. The “balanced force” concept for instrumentation of curved canals. J Endod 1985;11:203-11.
Yared G. Canal preparation using only one Ni-Ti rotary instrument: Preliminary observations. Int Endod J 2008;41:339-44.
Kim JW, Ha JH, Cheung GS, Versluis A, Kwak SW, Kim HC, et al.
Safety of the factory preset rotation angle of reciprocating instruments. J Endod 2014;40:1671-5.
Martinho FC, Gomes AP, Fernandes AM, Ferreira NS, Endo MS, Freitas LF, et al.
Clinical comparison of the effectiveness of single-file reciprocating systems and rotary systems for removal of endotoxins and cultivable bacteria from primarily infected root canals. J Endod 2014;40:625-9.
Bürklein S, Schäfer E. Apically extruded debris with reciprocating single-file and full-sequence rotary instrumentation systems. J Endod 2012;38:850-2.
Machado ME, Nabeshima CK, Caballero-Flores H, Elmadjian-Filho M, Duarte MA, Odinola-Zapata R, et al.
Instrument design may influence bacterial reduction during root canal preparation. Braz Dent J 2017;28:587-91.
Cavalli D, Toia CC, Flores Orozco EI, Khoury RD, Cardoso FG, Alves MC, et al.
Effectiveness in the removal of endotoxins and microbiological profile in primary endodontic infections using 3 different instrumentation systems: A randomized clinical study. J Endod 2017;43:1237-45.
Martinho FC, Freitas LF, Nascimento GG, Fernandes AM, Leite FR, Gomes AP, et al.
Endodontic retreatment: Clinical comparison of reciprocating systems versus rotary system in disinfecting root canals. Clin Oral Investig 2015;19:1411-7.
Ferrer-Luque CM, Bejarano I, Ruiz-Linares M, Baca P. Reduction in Enteroccocus faecalis
counts – A comparison between rotary and reciprocating systems. Int Endod J 2014;47:380-6.
Neves MA, Provenzano JC, Rôças IN, Siqueira JF Jr. Clinical antibacterial effectiveness of root canal preparation with reciprocating single-instrument or continuously rotating multi-instrument systems. J Endod 2016;42:25-9.
Rôças IN, Lima KC, Siqueira JF Jr. Reduction in bacterial counts in infected root canals after rotary or hand nickel-titanium instrumentation – A clinical study. Int Endod J 2013;46:681-7.
McGurkin-Smith R, Trope M, Caplan D, Sigurdsson A. Reduction of intracanal bacteria using GT rotary instrumentation, 5.25% NaOCl, EDTA, and Ca (OH) 2. J Endod 2005;31:359-63.
Huffaker SK, Safavi K, Spangberg LS, Kaufman B. Influence of a passive sonic irrigation system on the elimination of bacteria from root canal systems: A clinical study. J Endod 2010;36:1315-8.
Vera J, Siqueira JF Jr., Ricucci D, Loghin S, Fernández N, Flores B, et al.
One- versus two-visit endodontic treatment of teeth with apical periodontitis: A histobacteriologic study. J Endod 2012;38:1040-52.
Nair PN, Henry S, Cano V, Vera J. Microbial status of apical root canal system of human mandibular first molars with primary apical periodontitis after “one-visit” endodontic treatment. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2005;99:231-52.
Arias A, Perez-Higueras JJ, de la Macorra JC. Differences in cyclic fatigue resistance at apical and coronal levels of reciproc and waveone new files. J Endod 2012;38:1244-8.
Kakehashi S, Stanley HR, Fitzgerald RJ. The effects of surgical exposures of dental pulps in germ-free and conventional laboratory rats. Oral Surg Oral Med Oral Pathol 1965;20:340-9.
Aydin C, Tunca YM, Senses Z, Baysallar M, Kayaoglu G, Ørstavik D, et al.
Bacterial reduction by extensive versus conservative root canal instrumentation in vitro
. Acta Odontol Scand 2007;65:167-70.
Berutti E, Castellucci A. Cleaning and shaping the root canal system. In: Castellucci A, editor. Endodontics. 1st
ed., Vol. II. Firenze: IlTridente; 2005. p. 429-57.
Yang G, Yuan G, Yun X, Zhou X, Liu B, Wu H, et al.
Effects of two nickel-titanium instrument systems, Mtwo versus ProTaper universal, on root canal geometry assessed by micro-computed tomography. J Endod 2011;37:1412-6.
Machado ME, Sapia LA, Cai S, Martins GH, Nabeshima CK. Comparison of two rotary systems in root canal preparation regarding disinfection. J Endod 2010;36:1238-40.
Nabeshima CK, Caballero-Flores H, Cai S, Aranguren J, Borges Britto ML, Machado ME, et al.
Bacterial removal promoted by 2 single-file systems: Wave one and one shape. J Endod 2014;40:1995-8.
Dr. Malli Sureshbabu Nivedhitha
Department of Conservative Dentistry and Endodontics, Saveetha Dental College, Saveetha Institute of Medical and Technical Sciences, 162, Poonamallee High Road, Chennai - 600 077, Tamil Nadu
Source of Support: None, Conflict of Interest: None
[Figure 1], [Figure 2], [Figure 3]
[Table 1], [Table 2], [Table 3]
|This article has been cited by|
||Systemic inhibition of 5-lipoxygenase by MK-886 exacerbates apical periodontitis bone loss in a mouse model
| ||Igor Bassi Ferreira Petean, Alice Corrêa Silva-Sousa, Raquel Assed Bezerra da Silva, Marília Pacífico Lucisano, Léa Assed Bezerra da Silva, Guilherme Piedade Assed de Castro, Manoel Damião Sousa-Neto, Lúcia Helena Faccioli, Francisco Wanderley Garcia Paula-Silva |
| ||BMC Oral Health. 2023; 23(1) |
|[Pubmed] | [DOI]|
||An Appraisal on Newer Endodontic File Systems: A Narrative Review
| ||Shalini Singh, Kailash Attur, Anjali Oak, Mohammed Mustafa, Kamal Kumar Bagda, Nishtha Kathiria |
| ||The Journal of Contemporary Dental Practice. 2023; 23(9): 944 |
|[Pubmed] | [DOI]|
||Influence of needle working length and root canal curvature on irrigation: a computational fluid dynamics analysis based on a real tooth
| ||Na Zhou, Zhengqiu Huang, Mingzhou Yu, Shuli Deng, Baiping Fu, Hanhui Jin |
| ||BMC Oral Health. 2022; 22(1) |
|[Pubmed] | [DOI]|
||Economic analysis of the different endodontic instrumentation techniques used in the Unified Health System
| ||Laura Paredes Merchan, Livia Fernandes Probst, Ana Clara Correa Duarte Simões, Augusto Cesar Santos Raimundo, Yuri Wanderley Cavalcanti, Denise de Fátima Barros Cavalcante, João Victor Frazão Câmara, Antonio Carlos Pereira |
| ||BMC Oral Health. 2022; 22(1) |
|[Pubmed] | [DOI]|
||Centric versus eccentric engine-driven endodontic instrument design for microbial load reduction—A systematic review and meta-analysis of randomised clinical trials
| ||Jerry Jose, Kavalipurapu Venkata Teja, Emmanuel João Nogueira Leal Silva, Krishnamachari Janani, Riluwan Siddique, Giampiero Rossi-Fedele |
| ||Australian Endodontic Journal. 2022; |
|[Pubmed] | [DOI]|
||Influence of the type of reciprocating motion on the cyclic fatigue resistance of reciprocating files in a dynamic model
| ||Álvaro Zubizarreta-Macho, Alberto Albaladejo Martínez, Carlos Falcão Costa, Norberto Quispe-López, Ruben Agustín-Panadero, Jesús Mena-Álvarez |
| ||BMC Oral Health. 2021; 21(1) |
|[Pubmed] | [DOI]|
||In vitro evaluation of efficacy of two endodontic sonic-powered irrigant agitation systems in killing single-species intracanal biofilms
| ||Chang Zeng, Joseph Everett, Stephanie Sidow, Brian E. Bergeron, Fucong Tian, Jingzhi Ma, Franklin R. Tay |
| ||Journal of Dentistry. 2021; 115: 103859 |
|[Pubmed] | [DOI]|
||Accuracy of different generations of apex locators in determining working length; a systematic review and meta-analysis
| ||Kaveh Nasiri, Karl-Thomas Wrbas |
| ||The Saudi Dental Journal. 2021; |
|[Pubmed] | [DOI]|
||Evaluation of Dentinal Crack Formation by ProTaper Next, HyFlex CM and Wave One Reciprocating File Systems at the Apical and Middle Third of Root in Mandibular Molars - An In-Vitro Scanning Electron Microscope Analysis
| ||Gis George, Ranjini M. A, Swapna D. V, Rashmi K, Jesline Merly James, Roopa R. Nadig |
| ||Journal of Evolution of Medical and Dental Sciences. 2021; 10(32): 2597 |
|[Pubmed] | [DOI]|
||The impact of Distinctive Root Canal Instrumentation Systems on Endotoxin Lessening from the Root Canal: A Systematic Review and Meta-Analysis
| ||Navid Nasrabadi, Samira Jamali, Marzieh Ghasemi Vojoodi, Marzieh Jamali |
| ||Pesquisa Brasileira em Odontopediatria e Clínica Integrada. 2021; 21 |
|[Pubmed] | [DOI]|
||Factors influencing apical debris extrusion during endodontic treatment - A review of the literature
| ||Natasa Predin Djuric, Peet J Van der Vyver, Martin Vorster, Zunaid I Vally |
| ||South African Dental Journal. 2021; 76(1): 28 |
|[Pubmed] | [DOI]|
||Evaluation of preparation times of WaveOne Gold reciprocating instruments compared to two analogous counterparts
| ||Christiaan Victor, Peet J Van der Vyver, Martin Vorster, Farzana Paleker, Zunaid I Vally |
| ||South African Dental Journal. 2021; 76(06): 315 |
|[Pubmed] | [DOI]|
||The Effect of Taper and Apical Diameter on the Cyclic Fatigue Resistance of Rotary Endodontic Files Using an Experimental Electronic Device
| ||Vicente Faus-Llácer, Nirmine Hamoud Kharrat, Celia Ruiz-Sánchez, Ignacio Faus-Matoses, Álvaro Zubizarreta-Macho, Vicente Faus-Matoses |
| ||Applied Sciences. 2021; 11(2): 863 |
|[Pubmed] | [DOI]|
||Influence of reciprocating and rotary instrumentation on microbial reduction: a systematic review and meta-analysis of in vitro studies
| ||Selen Küçükkaya Eren, Emel Uzunoglu-Özyürek, Sevilay Karahan |
| ||Restorative Dentistry & Endodontics. 2021; 46(2) |
|[Pubmed] | [DOI]|
||The effectiveness of the supplementary use of the XP-endo Finisher on bacteria content reduction: a systematic review and meta-analysis
| ||Ludmila Smith de Jesus Oliveira, Rafaella Mariana Fontes de Bragança, Rafael Sarkis-Onofre, André Luis Faria-e-Silva |
| ||Restorative Dentistry & Endodontics. 2021; 46(3) |
|[Pubmed] | [DOI]|
||Current technology in endodontic instrumentation: advances in metallurgy and manufacture
| ||Faith Campbell, Joanne Cunliffe, James Darcey |
| ||British Dental Journal. 2021; 231(1): 49 |
|[Pubmed] | [DOI]|
||Root canal preparation with reciprocating instruments - A literature review and clinical application
| ||Christiaan Victor, Peet J Van der Vyver, Martin Vorster, Zunaid I Vally |
| ||South African Dental Journal. 2020; 75(9): 493 |
|[Pubmed] | [DOI]|
||Comparison of antibacterial effectiveness of three rotary file system with different geometry in infected root canals before and after instrumentation–a double-blinded randomized controlled clinical trial
| ||Riluwan Siddique, Malli Sureshbabu Nivedhitha, Manish Ranjan, Benoy Jacob, Pradeep Solete |
| ||BDJ Open. 2020; 6(1) |
|[Pubmed] | [DOI]|
| Article Access Statistics|
| Viewed||12146 |
| Printed||383 |
| Emailed||0 |
| PDF Downloaded||645 |
| Comments ||[Add] |
| Cited by others ||18 |